For more information and discussion, please go and visit the buyer forum
— History Versions —
Reading the manual is the first priority
How to monitor the voltage and current between type-C devices?
1. Make sure the PD trigger switch on the meter is turned off.
2. Connect the charger and the meter with a type-C cable, and then connect the meter and the mobile phone (or something else) with another type-C cable.
3. If the screen didn’t light up, flip the cable on any side of the meter.
–Voltage and current monitoring with 0.0001V, 0.0001A resolution, 0.05% precision.
–Interfaces supported for both USB-A and USB Type-C devices.
–Voltage and current curve painting with sample rate up to 100 samples per second.(Note: Curve painting mode only, recording cannot be done at that rate)
–5 groups of capacity, energy, time recording.
–D+ D- voltage measure with 0.01V resolution, 2% precision.
—PD2.0, PD3.0 with PPS (Programmable Power Supply) trigger supported
—QC2.0 (5V/9V/12V/20V), QC3.0, QC4.0 (Or PPS) trigger supported.
—Samsung AFC 9V, Huawei FCP 9V, Huawei SCP 4.5V trigger supported.
—Apple 2.4A trigger supported.
—Apple MFI cable authorization test (Reference only).
—Protocol recognize for QC2.0-5V/9V/12V/20V, QC3.0, Apple 0.5A/1A/2.1A/2.4A, USB-DCP, Huawei FCP/SCP, Samsung AFC.
—Auto enumerate protocols function for chargers/power banks.
—Offline curve recording with 4 datasets, 3750 data points per set. Programmable recording rate, record start/stop threshold. Curves can be uploaded to PC software for further analysis.
— Power Delivery protocol listener (or analyzer) with full packets analysis for PD developers. PC software provided.
— Cable ESR test supported for Type-A to Type-C and Type-C to Type-C cables.
— 4-line ESR test method software supported, accessory not provided.
–Firmware can be updated by PC software directly, for PC software (with corresponding firmware inside), please visit official website of AVHzY.
–One temperature sensor on board with 0.1C resolution.
–One calendar/digital clock, can be used when the device is not occupied.
Main Control IC：STM32 F072
Display screen：TFT 1.44″ 128*128
Voltage Current Scale ：DC 4-26V 0-6A (0.1-26V with independent micro USB supplied)
Voltage Current Resolution：0.0001V 0.0001A
Voltage Current Accuracy：0.1%+2d
Current Sample Point：High-side
Overall Internal Resistance(Type-A & Type-C)：28mOhm （After 1000 cycles）
Quiescent Current：0.4mA （At DC 5V）
Electronic Load Parameters:
Applied voltage range: DC 3.7V-13V
Adjustable current range: 0.15A-3.00A
Electricity discharging: continuous 15W
Temperature control: about 40 ℃
1 X USB Meter,1 X USB Electronic Load,1 X Quick Tour.